REED ST-156 Features
- Dual technology provides automatic recognition for ferrous and nonferrous metal substrates
- Magnetic induction for ferrous metal substrates
- Eddy current measurement for non-ferrous metal substrates
- One or two point calibration
- User-settable high/low alarms
- Easy Zero calibration
- Min, Max, and Average functions
- Memory stores up to 400 readings
- Easily delete single or group readings
- Auto-Power-off function Backlight
- Measurement Range: 0 to 1250 μm (0 to 49 mils)
How to use Reed ST-156 Coating Thickness Gauge
Continuous Measuring Mode
In Continuous Mode, the probe is not lifted off the surface being measured. Readings are not accompanied by a beep. All readings taken in this mode are automatically entered into the statistics program as long as sufficient memory is available.
Single Measuring Mode
In Single Mode, the probe is lifted off the surface being measured and moved. Readings are accompanied by a beep. All readings taken in this mode are automatically entered into the statistics program as long as sufficient memory is available.
The probe can work in three modes:
AUTO: The probe can automatically select the working mode. When placed on steel (magnetic substrates), it will work in magnetic induction principle. When placed on no-ferrous metals, it will work in eddy current principle.
Fe: The probe works in magnetic induction principle.
No-Fe: The probe works in eddy current principle.
While in the Menu, choose “Statistic View” to toggle between Average, Maximum, Minimum and Standard Deviation functions. Press the Red Bar Button to select the current Statistic View and to return to measurement mode. A corresponding symbol of the Statistic View will be displayed on the lower right of the LCD. At the same time, the statistic number of the measured readings will display on the lower left of the LCD. Via the “Statistic View” in the Menu, you can look through all statistic values for a current group.
In the Menu, choose “Measurement View” to look through all measurement readings for current group.
REED ST-156 Specifications
|Measuring Range||0 to 1250µm
0 to 49.21mils
|Accuracy||0 to 850µm: ±3% +1µm
850 to 1250µm: ±5%
0 to 33.46mils: ±3% +0.039mils
33.46 to 49.21mils: ±5%
|Resolution||0 to 50µm: 0.1µm
50 to 850µm: 1µm
850 to 1250µm: 0.01mm
0 to 1.968mils: 0.001mils
1.968 to 33.46mils: 0.01mils
33.46 to 49.21mils: 0.1mils
|Min. Curvature Radius||F: 1.5mm
|Min. Area Diameter||F: 7mm
|Min. Substrate Thickness||F: 0.5mm
|Power Supply||2 x 1.5V “AA” batteries|
|Dimensions||110 x 50 x 23mm|