Tenmars TM-508A Milli-Ohmmeter Specifications:
|Measuring Ranges||Accuracy||Resolution||Test Current||Open-Terminal Voltage|
|200 milliohms||±(0.3% of rdg + 1dgt)||0.1 milliohm||100mA||1.0 Volt maximum|
|2 ohms||1 milliohm||10mA|
|20 ohms||10 milliohms||10mA|
|200 ohms||0.1 ohm||1mA|
|2k ohms||1 ohm||1mA|
|20 kohms||10 ohms||100µA|
Tenmars TM-508A Battery Replacement
The meter is powered by 1.5V battery x6pcs (NEDA 15F IEC R6 JIS UM-3). Use the following procedure to replace the battery:
1. Replace battery immediately when LCD displays “ ”.
2. Turn the function switch to the “OFF” position.
3. Use a screwdriver to unfasten the screws on the battery cover and remove the cover.
4. Take out the old batteries and replace with new batteries, taking care to note the correct polarity.
5. Re-install the battery cover and tighten the holding screws.
Replacement of fuse
For a safety protection of the electrical circuit please replace new fuse complying the specification 0.5A/250V, 5Øx 20mm. FAST MIN INTERRUPT RATINGS 1500A.
1. Select the function range switch to “OFF” position. To unscrew the bottom case by using a screwdriver.
2. Pull out the burned fuse and replace a new one.
3. Close the bottom case and tighten screw.
Maintenance and Care
1. That all necessary requirements of inspection and maintenance are not mentioned in this manual, a qualified technician should perform it.
2. This meter is a precision digital instrument, whether in use or in storage, please do not exceed the specification requirements to avoid any possible damage or danger during use.
3. Do not use strong or abrasive detergents, water and wet cloth to clean the instrument. Do use a dry cloth to clean the instrument.
4. Do not place this meter in high temperature or humidity or expose to direct sunlight.
5. Once the measurement is completed, turn the rotary switch to OFF, Remove the batteries from battery holder if the instrument is not is used for a long period in order to avoid the liquid leakage from the battery.